I have lost my colour chart. Can I download the colour chart for the test?
The colour cards are not available for download on the JBL homepage. For this a guaranteed true colour print is required, and you need a printer which can be re-calibrated before each printout. Most privately used printers are not accurate enough for this.
We will happily send you a new colour chart bundle with all individual colour charts for the JBL tests free of charge. Please forward the JBL customer service your address.
What can I do when the measured value of a water test falls outside the scale?
If one of your measurements reaches the end of the colour chart scale you can multiply the scale, by diluting the following tests with distilled water (e.g. JBL Dest).
Here’s how to do it: dilute the sample with distilled water and carry out the test once more.
Her’s how in greater detail:
JBL GH test kit: Instead of 5 ml use 10 ml aquarium water. Now 1 drop is equivalent to 0.5 dGH.
JBL KH test kit: Instead of 5 ml use 10 ml aquarium water. Now 1 drop is equivalent to 0.5 dGH.
Note: when using 10 ml sample water the colour development is weaker with the first drops than with 5 ml sample water. Put the measuring container on a white underlay for a better readability.
JBL NH4/NH3 Ammonium/Ammonia test kit, JBL NO2 Nitrite test kit, JBL Fe Iron test kit and JBL Cu Copper test kit:
2.5 ml sample + 2.5 ml dist. water: result times 2
1 ml sample + 4 ml dist. water: result times 5
0.5 ml sample + 4.5 ml dist. water: result times 10
NO3 Nitrate test kit, JBL SiO2 Silicate test kit, JBL PO4 Phosphate sensitive test kit and JBL Mg Magnesium test kit freshwater:
5 ml sample + 5 ml dist. water: result X 2
2 ml sample + 8 ml dist. water: result X 5
1 ml sample + 9 ml dist. water: result X 10
JBL PO4 Phosphate test kit koi high range:
2.5 ml sample + 2.5 ml dist. water: result X 2
1 ml sample + 4 ml dist. water: result X 5
JBL K Kalium test kit:
7.5 ml sample + 7.5 ml dist. water: result X 2
3.75 ml sample + 12.25 ml dist. water: result X 5
1.5 ml sample + 14.5 ml dist. water: result X 10
JBL CO2 direct test kit, JBL Ca Calcium test kit and JBL Mg+Ca Magnesium+Calcium test kit:
Which of the three scales on the colour cards symbolize of the water tests saltwater, freshwater and garden pond ?
koi carp = garden pond
angel fish = saltwater
discus = freshwater
Which scale applies for my aquarium? My test has two or three control fields with fish and recommendations.
We repeatedly receive questions regarding the symbols on the water test. The hobby areas cannot be clearly assigned to the scales for analysis. There are three symbols to make the hobby areas on the colour charts clear:
Discus = fresh water
Emperor = salt water
Koi = garden pond
In the following, you will find a list which assigns the scale for analysing the results from measurements in the hobby area and explains the symbols: The scales are described from the top to the bottom here.
NO3: salt water, fresh water, garden pond
pH 6.0 – 7.6: fresh water, garden ponds
pH 7.4 – 9.0: salt water, fresh water, garden pond
Fe: fresh water, garden pond
The following JBL tests have a uniform scale for all 2 or 3 hobby areas for analysis of the results from the measurements, as differentiation is no longer necessary:
My nitrate test indicates measurement results which I can’t read or which are far too low.
The reason is often an incorrect use of the test. It is important to shake the test vigorously for precisely one minute. All the other important steps can be read in the instructions.
The instructions point out that the comparator block should be used to compare the colours. Can this be done without the comparator block as well ? Or is there a factor from this comparison that is required for the measurement ?
The colour charts are intended to be used with the comparator block. In addition, when the comparator block is used with the blank water sample, this compensates the inherent water colour.
Therefore, if the comparator block is not used, there will most likely be deviations in the readings.
When doing the nitrate test, why can't I use 5 ml of water, one spoon of Reagent 1 and 3 drops of Reagent 2 ? Wouldn't that result in twice as many measurements ?
If you use 5 ml instead of the 10 ml of sample liquid, this would yield distorted results for two reasons.
For one, the powder for the reduction of NO₃ doesn't dissolve completely so that it forms a thin layer of sediment which causes difficulties when reading the results with too little water. Tapping the cuvette on the side several times helps by causing the powder to be deposited on the side.
Secondly, attention must be paid to the drop size of reagent 2, as the first drop from the drop dispenser can turn out to be smaller in individual cases, especially when the bottle is not held in a fully vertical position. This may also lead to incorrect measuring results. Therefore, it is advisable to use 10 ml in order to ensure accurate test results. Please make sure to really shake hard for exactly one minute in order for the colour to develop properly.
What is the shelf life of the JBL Nitrate Test haltbar ?
The JBL nitrate test has a shelf life of 2 years in the original sealed bottle.
The following is printed on the bottom of the packaging:
CH/B: 2262 (e.g.) = batch number
BBD: 05/12 (e.g.) = best-before date (month/year)
Please note that the best-before date does not apply for opened reagents. Exposure to atmospheric oxygen decreases the shelf life.
Do my water tests still work although they have been open for nine months?
Basically a sealed test keeps longer than an open one. How much the shelf life is reduced depends crucially on the handling and the storage conditions of the reagents. It may happen that a reagent in a test that has been open for 9 months doesn’t indicate correctly any more. However, additional information is needed to answer this question properly.